Strain Gauge Accelerometer – Diagram and Working
How Accelerometer Works
An accelerometer Sensor is an electromechanical device that measures acceleration forces. These forces may be static or dynamic which is caused by moving or vibrating the accelerometer. By measuring the amount of static acceleration due to gravity, you can find out the angle the device is tilted at with respect to the earth. By sensing the amount of dynamic acceleration, you can analyze the way the device is moving.
There are different types of accelerometers:
- Piezo electric accelerometer
- Piezo resistive accelerometer
- Strain gage accelerometers
Strain Gauge Accelerometer
Basic Principle of Strain Gauge Accelerometer
When a cantilever beam attached with a mass at its free end is subjected to vibration, vibrational displacement of the mass takes place. Depending on the displacement of the mass, the beam deflects and hence the beam is strained. The resulting strain is proportional to the vibration displacement of the mass and hence the vibration/acceleration being measured when calibrated.
Description of Strain Gauge Accelerometer
The main parts of a strain gague accelerometer are as follows:
- A cantilever beam fixed to the housing of the instrument.
- A mass is fixed to the free end of the cantilever beam.
- Two bounded strain gauges are mounted on the cantilever beam as shown in diagram.
- Damping is provided by a viscous fluid filled inside the housing.
Operation of Strain Gauge Accelerometer
- The accelerometer is fitted on to the structure whose acceleration is to be measured.
- Due to the vibration, vibrational displacement of the mass occurs, causing the cantilever beam to be strained.
- Hence the strain gauges mounted on the cantilever beam are also strained and due to this their resistance change.
- Hence a measure of this change in resistance of the strain gauge becomes a measure of the extent to which the cantilever beam is strained.
- But the resulting strain of the cantilever beam is proportional to the vibration/acceleration and hence a measure of the change in resistance of the strain gauges becomes a measure of vibration/acceleration.
- The leads of the strain gauges are connected to a wheat stone bridge whose output is calibrated in terms of vibration/acceleration.
Advantages of Strain gauge accelerometer
- Strain gage accelerometers are preferred in high shock applications.
- They can measure accelerations down to zero Hertz.
Disadvantages of Strain gauge accelerometer
- Limited high frequency response
More Resources /articles
Measurement Science and Metrology Notes , Articles
Mechatronics System Notes , article , Interview Que and Ans
Mechanical Subjectwise Basic Concept Notes ,Articles
Simple , Easy mechanical projectsManually Hand Operated projectsAffordable low budget Projects
Mechanical engineering is one of the most physically demanding professions that involve designing, manufacturing, and maintaining mechanical systems. A mechanical engineer's work involves utilizing...
Management is an essential component of every project and team. A competent manager is indispensable in guiding the team, owning outcomes, and mediating conflicts. Engineering managers, in...