Eddy current sensors | Advantages ,Disadvantages and Application
Eddy current sensors detect the proximity or presence magnetic fields generated by a reference coil. Eddy current non-ferrous metals. They can be used as proximity sensors of a target by sensing the sensors detect ferrous and to detect the presence of gauges target or it can be configured to measure the position or displacement of a target.
An eddy current is a local electric current induced in a conductive material due to the magnetic field produced by the sensor or active coil. It is sensed by a reference coil to create an output signal. When the distance changes between target and probe, the impedance of the coil will correspondingly change. The change in impedance can be detected by a carefully arranged bridge circuit shown in Figure 1.36.
The eddy currents are confined to shallow depths near the conductive target
surface. Their effective depth is given by
The target material must be at least three times thicker than the effective depth of the eddy currents to make the transducer successful because the transducer assumes that the eddy currents are localized near the surface of a semi-infinite solid and the actual eddy current amplitude decreases quadratically with distance.
Applications of Eddy current sensors
Since it is a non-contact device, it is suitable for higher resolution measurement applications. The device is used for finding out the position of an object that is conductive in nature. Various applications are as follows.
(i) Position measurement
Since the output of an eddy current transducer represents the size of the distance between transducer and conductor, the device can be calibrated to measure the position or displacement of the target. Thus, it can be applicable in monitoring or sensing the precise location of an object such as a machine tool. It can also be used to locate the final position of precise equipment such as a disk drive.
(ii) Vibrating motion measurement
The device is also suitable for finding the alternate positions of a vibrating conductor. Since a contact device is impracticable for this application, a non-contact device such as eddy current transducer is highly recommended. Thus, it can be applicable in measuring the distance of a shaft from a reference point or the to-and- fro movement of vibrating instruments.
Advantages of Eddy current sensors
(i) It is compact in size
(ii) Cost is low.
(iii) Reliability is high.
(iv) It produces high frequency response.
(v) Sensitivity for small displacement is high.
(vi) It is insensitive to material in the gap between transducer and conductor.
Disadvantages of Eddy current sensors
(i) The result will be precise only if the gap between transducer and conductor is small.
(ii) The device cannot be used for finding the position of non-conductive materials.
(iii) There always occurs a non-linear relationship between distance and impedance of the active coil of the device.
(iv) The device is highly temperature sensitive.
Introduction to Pressure Vessels Vessels, tanks, and pipelines that carry, store, or receive fluids are called pressure vessels. A pressure vessel is defined as a container with a pressure...
Knuckle Joint A knuckle joint is used to connect two rods which are under the action of tensile loads. However, if the joint is guided, the rods may support a compressive load. A knuckle joint...